The plugin requires a valid Dewesoft license. To test the plugin, you can use an Evaluation license. For more information about licensing follow this LINK.
Simply extract the contents of OpcUaClient.zip archive into the Addons folder of your Dewesoft installation. (e.g., C:\DEWESoft\Bin\X3\Addons). Then you can start Dewesoft as an administrator and register the plugin (aka. extension). Click Options, Settings, select Extensions and click the plus sign. Then find the OPC UA Client plugin in the list and activate it (i.e. click the check-box in Figure4) - when the plugin does not show up in the list, you may need to register it first.
Registering the plugin
Before you can use Plugins in Dewesoft, they must be registered once. When Dewesoft is started it will try to register all plugins (*.dll files) that it finds in the Addons folder. In order to do that, DEWESoft® requires administrator permissions (because it must write to the Windows® registry). When Dewesoft is not run with administrator permissions, the registration cannot be done automatically. When the plugin does not show up in the Extensions list, you must press the Refresh button (Figure 5 - Update extensions list). Note: you may need to start Dewesoft as administrator (depending on the UAC settings of your Windows user/installation). When you have pressed the Refresh button, the registration Window will be shown for a short time (Figure 6). After that, you need to restart Dewesoft.
In the Extensions click on the OPC UA Client, then on the right the settings for the plugin will open. Here you can select the previous mentioned operation mode (clock mode). To add a server to connect to, you need to click on the “+” and enter the URL and optionally the Name. After this is done you can exit the settings by clicking “OK”.
OPC UA Client setup screen
The OPC UA Client setup screen allows you to select the Server by clicking the dropdown menu. In Config tabyou can use the browser to browse through the information model. The current location is shown on top of the Browser. You can write the location path and click on Go to jump in the location. Under Nodes you can see nodes contained at chosen path along with properties: Identifier, Name, Type and Status of each nod. You can select which nodes you want to use in Dewesoft with the Used toggle (Off/On) button. In the Channels tab (Figure 9) you can see all the used OPC UA channels. OPC UA method nodes and grouped as Methods, OPC UA variables and properties are grouped as Variables. For each channel you can see different information and attributes. Under Type you can see the data type of the variable. If “DsChannelType” is shown here then Dewesoft DAQ profile mode is enabled and data type is inherently Double. With Sample rate you can choose the rate at which data is being fetched from the server. If this attributeis read-only then it shows the original sampling rate at the source (Dewesoft OPC UA Server). Throughput shows incoming rate (samples per second) and this depends on network bandwidth. Values represents last node value and can give quick insight if channel us alive. With Units you can set the channels units. The OPC UA Client plugin supports Methods with input and output parameters. The following section provides more details on this topic.
The OPC UA Client plugin supports calling OPC UA methods. You can easily associate method call, input or output with various Input control display visual controls and design a nice customized user interface. Calling method is normally associated to push button.To do so go to design mode, add “Input control display”, choose “Control channel” as Display type and pick “Push button” form the dropdown menu. You can also enter custom Button text. On the right select channel “Channel”. To use input or output parameters you can add another input control display, choose “Control channel” as Display type and pick “Input field” or other suitable type. On the right select channel containing either “outparam” or “inparam”(Figure 10). Leave design mode, enter some parameters into your input control, press enter to confirm input and then click on the button and see the results in output associated control.
With the Input control display we can also remotely change the values of the OPC UA variables. Variables should have writable permission on the server side. In design mode add Input control display, choose “Control channel” as Display type, from the dropdown menu choose Input field or any other suitable type (turn knob, slider, etc.). Then click the desired channel on the right. Leave design mode to preform control.